[Mandible single case] When Ø5.0 stent cannot be fabricated due to narrow distance between the teeth.
[Mandible multiple case] When fixture cannot be place near adjacent teeth due to large stent core on regular stent.
Initial Drill • Use the initial drill in order to mark the drilling position on the bone. Start drilling slowly, when the drill guide part is fully contacted with the drilling core of R2GATE Guide™. • Recommended drilling speed range is 300 ~ 800 RPM with copious irrigation. Second Drill • This unique step-drill(from Ø2.0 to Ø4.6) is used to flare out the upper cortical bone of the osteotomy. • It helps not only the rest drilling procedure but abutment connection. In case of hard bone, if the 2nd drilling will be disturbed by thick cortical bone. Stop the drilling and try it after the final drilling procedure.
Stopper Drill • Universal drills consist of Ø2.0, Ø.2.5, Ø2.8 diameter to enlarge the osteotomy gradually. • The length of the drill are designed as 7.0, 8.5, 10, 11.5,13mm for the most common length of the implant system. • Recommended drilling speed range is 500 ~ 800 RPM with copious irrigation.
Tap Drill • The purpose of tab drills in the universal kit system is insertion test. • Recommended insertion torque and speed is 45 ~ 50Ncm, under 40 RPM. Implant Carrier • To pick up the fixture from the ampule and insert it into the osteotomy. Then turn it to clock-wise direction 2~3 times manually. • When it gets fixation from the osteotomy, connect the handpiece adaptor and use the implant motor. • Recommended insertion torque is 45~50Ncm.
Sign up for our newsletter
Sign up for our mailing list to receive new product alerts, special offers, and coupon codes.